1200℃Single temperature zone tube furnace
From the details, after more than a hundred times of repeated refining, so as to create this delicate and beautiful appearance and a 7 inch touch screen operating system tube furnace.It is mainly used in the fields of crystal annealing, electronic lighting, 2-d material sintering, atmosphere protection sintering, vacuum coating, CVD experiment, material composition determination and so on.
|1. Double air-cooled structure;
n2. High purity Al2O3 fiber refractory insulation material, energy saving and consumption reduction;
n3. The high purity alumina coating on the heating element can prolong the service life;
n4. NBD101 intelligent control system, let you worry-free operation, 3 minutes to get started.
|Electrical specifications||AC220V 4KW|
|Highest temperature||1200 ℃ （<1H）|
|Working temperature||1100 ℃|
|Maximum heating rate||≤ 20 ℃/min|
|Temperature zone length||440mm|
|Furnace pipe size||Φ50*1000mm||Φ60*1000mm||Φ80*1000mm||Φ100*1000mm|
|Temperature field distribution||As shown in the figure: here is the temperature curve of the equipment (size of the constant temperature zone: 260mm).|
|Control system||1. 15 temperature curves can be stored in advance to avoid the trouble caused by repeated Settings of different experimental processes;
n2. The experimental process is more intuitive and the operation is more convenient;
n3. Wireless control module can be selected to realize remote control;
n4. It has the functions of over temperature alarm, broken couple prompt, leakage protection and so on.
|Temperature precision||+/- 1 ℃|
|Heating element||Mo – doped fe-cr-al alloy|
|Vacuum degree: ≤10Pa (mechanical pump)|
|Pressure measurement and monitoring||Mechanical pressure gauges with damping oil are used to reduce the back and forth swing of the pointer.|
|Gas supply system||The float flowmeter is used to control the gas velocity, which is integrated with the equipment and has been tested for air leakage before delivery.|
|The net weight||60KG|
|Precautions for equipment use||1. When the furnace temperature of the equipment is ≥300℃, do not open the furnace to avoid damage;
n2. When the equipment is in use, the pressure inside the furnace shall not exceed 0.125mpa (absolute pressure) to prevent damage caused by excessive pressure;
n3. When used under high vacuum (10 to the minus 3 Pa), the operating temperature of the equipment shall not exceed 800℃.
|Service support||1 year warranty with lifetime support (warranty excludes consumable parts such as furnace tubes and o-rings)|