1200℃ Intelligent tube furnace
We continue to start with the details and after more than 100 times of repeated tempering, we have created this exquisite and beautiful tube furnace with a 7-inch touch screen operating system. This product is mainly used in the fields of crystal annealing, electronic lighting, two-dimensional material sintering, atmosphere protection sintering, vacuum coating, CVD experiment, material composition determination, etc. It is trusted by researchers in various fields.
|1. Double-layer steel structure;
n2. High-purity Al2O3 fiber refractory insulation material, maximum energy saving;
n3. High-purity alumina coating on refractory ceramics can prolong the service life;
n4. Alumina tube plug can help you establish a custom temperature gradient in the tube.
|Electrical Specifications||AC220V 4KW|
|Max temperature||1200 ℃ （<1小时）|
|Operating temperature||1100 ℃ （连续）|
|Heating rate||≤ 20 ℃/min|
|Heating zone length||440mm|
|Furnace tube size||Φ50*1000mm||Φ60*1000mm||Φ80*1000mm||Φ100*1000mm|
|Temperature distribution||As shown in the figure: this is the temperature curve of the device (constant temperature area: 260mm)|
|Control System||1. 15 temperature curves can be pre-stored to avoid the trouble caused by repeated setting of different experimental processes;
n2. The experimental process is more intuitive and the operation is more convenient;
n3. Optional wireless control module is available for remote control;
n4. With over-temperature alarm, broken couple reminder, leakage protection and other functions.
|Temperature control accuracy||+/- 1 ℃|
|Heating element||Mo-Fe-Cr-Al alloy|
|Vacuum: ≤10Pa (mechanical pump)|
|Pressure measurement and monitoring||A mechanical pressure gauge with shock-absorbing oil is used to dampen the pressure fluctuations and reduce the pointer swinging back and forth.|
|Gas supply system||The float flow meter is used to control the gas flow rate, which is integrated with the equipment, and the leak test has been carried out before leaving the factory.|
|Notes on equipment use||1. When the furnace temperature of the equipment is ≥300 ℃, it is forbidden to open the furnace to avoid injury;
n2. When the equipment is used, the pressure in the furnace tube must not exceed 0.125MPa (absolute pressure) to prevent the equipment from being damaged due to excessive pressure;
n3. When used under high vacuum (10 minus 3 power Pa), the temperature of the equipment should not exceed 800 ℃.
|Service support||1 year warranty with lifetime support (consumable parts are not included in the warranty, such as furnace tubes and O-rings)|